|FOB Unit Price:||US $10 US $8|
|Purchase Qty. (Pieces)||FOB Unit Price|
|Transport Package:||by Veneer Board Box|
|Payment Terms:||L/C, T/T|
- Install: Vertical
- Level Measurement: B
- Application: Civil
- Medium Pressure: General Water Meter
- Stucture: Wireless Communicating
- Usage: Water
- Size: 15mm-50mm
- Specification: 15mm-50mm
- Medium Temperature: Cold Water
- Counter Instructions: Digital and Analogue
- Principle: Volumetric Meter
- Work Environment: Dry Water Meter
- Structure: Brass Body; Plastic Cap or Option
- Production Standard: ISO 4064 Class C
- Trademark: No Brand
- Origin: China
Customer Question & Answer (0)
It is a multi-jet super dry water meter for residential application in sizes 15mm-50mm for cold meter.
Application: Used for measuring the total volume of water flowing through the vertical running water pipeline.
Features: This meter is a type of vertical installation, and it is characterized by advanced structure, straight reading, and accurate measurement, etc.
Water temperature: 0º C~40 º C for cold potable water
0º C~90 º C for hot water
Water pressure: ≤ 1Mpa
A) In the lower zone from qmin inclusive up to but excluding qt is ± 5%.
B) In the upper zone from qt inclusive up to and including QS is ± 2%
For cold potable water meter (± 3% for hot water meter).
Magnetic Drive, lower transmission resistance
Magnetic Shield , for external magnetic field protection
Sealed dry dial register ensures clear reading
Internal strainer, inlet strainer for selecting
External regulating device
Directive 2004/22/EC on measuring instruments; the European Standard EN14154+A1+A2: Water Meter; the International Organization of Legal Metrology OIML R-49: 2006(E)
Several lengths and connections available on request
Non return Valve
Reed switch option
Water temperature: ≤50ºC for cold water meter
Water pressure: ≤1MPa or 1.6MPa optional
(10bar or 16bar option)
The meter should be installed in horizontal position with the direction of the flow as
indicated by the arrow cast in the meter body with the register face upwards.
Pipeline must be flushed before installation.
The meter should be constantly full of water during operation.
MAXIMUM PERMISSIBLE ERROR
In the lower zone from Qmin inclusive up to but excluding Qt is ±5%.
In the upper zone from Qt inclusive up to and including Qs is: cold water meter ±2%.
Each water meter has one set couplings optional: 2pcs tube, 2pcs nut and 2pcs gasket